|Radcal SRC-500 Ionization Chamber is a direct replacement for the Fluke® 96010A Ion Chamber.
Radcal SRC-500 Ionization Chamber is having the same form factor but manufactured by Radcal Corporation it can be used in the Fluke® Radiation Leakage Detection System. This chamber meets 21 CFR 1020.30(k) for leakage measurement requirements* (Electrometer not provided)
* (k) Leakage radiation from the diagnostic source assembly. The leakage radiation from the diagnostic source assembly measured at a distance of 1 meter in any direction from the source shall not exceed 0.88 milligray (mGy) air kerma (vice 100 milliroentgen (mR) exposure) in 1 hour when the X-ray tube is operated at the leakage technique factors. If the maximum rated peak tube potential of the tube housing assembly is greater than the maximum rated peak tube potential for the diagnostic source assembly, positive means shall be provided to limit the maximum X-ray tube potential to that of the diagnostic source assembly. Compliance shall be determined by measurements averaged over an area of 100 square cm with no linear dimension greater than 20 cm.